Welcome to the fluidfm forum
This community is for professionals and enthusiasts of our products and services. Share and discuss the best applications, experiments and new ideas, build your professional profile and become a better researcher together.
Please read the guidelines before participating in this community.
Plasma treatment cleans your probe or sample surfaces and makes them more hydrophilic. This can significantly improve the quality of any coating and make filling of the probe faster and easier.
We recommend the following plasma treatment:
1. Place the probe with the blister pack into a plasma cleaner (remove the cover first).
2. Air plasma clean the FluidFM probe for 30 sec to 2 min at 18W radio frequency power ("High" - Harrick PDC 32 G).
The power and time values must be set up accordingly for every device.
Please try to give a substantial answer. If you wanted to comment on the question or answer, just use the commenting tool. Please remember that you can always revise your answers - no need to answer the same question twice. Also, please don't forget to vote - it really helps to select the best questions and answers!
About This Community
|Asked: 3/20/19, 8:29 AM|
|Seen: 41 times|
|Last updated: 5/14/19, 12:40 PM|