Welcome to the fluidfm forum
This community is for professionals and enthusiasts of our products and services. Share and discuss the best applications, experiments and new ideas, build your professional profile and become a better researcher together.
Please read the guidelines before participating in this community.
Our FluidFM cantilevers are made of silicon nitride and based on microfabrication technology. As shown in the picture below (a), they have a length of 200 µm and a width of 36 µm. The microchannel within the cantilever has a height that oscillates between 0.5 and 1.5 µm. At the far end of each cantilever, there is an aperture which connects the microchannel inside. Its size varies between 100 nm and 8 µm and gives, together with the tip shape, the functionality to the cantilever: nanopipette, nanosyringe or micropipette. In the case of the nanosyringes, the cantilever has a pyramid on the tip, with a height of 7 µm.
Figure. Scanning electron micrographs of FluidFM cantilevers (nanosyringes), showing their dimensions. (a) top view of a FluidFM cantilever, with a length and width of 200 µm and 36 µm respectively; (b) FIB cut through a nanosyringe shows the inner geometry of the pyramid (height 7µm) and the microchannel (height 0.5-1.5 µm).
More information about probe geometry can be found in the following entries:
About This Community
|Asked: 1/16/20, 10:35 AM|
|Seen: 82 times|
|Last updated: 1/17/20, 10:53 AM|