Welcome to the fluidfm forum
This community is for professionals and enthusiasts of our products and services. Share and discuss the best applications, experiments and new ideas, build your professional profile and become a better researcher together.
Please read the guidelines before participating in this community.
In order to avoid contamination in manipulated cell cultures, it might be necessary to sterilize the FluidFM probe. The protocol below was tested for a FluidFM micropipette (8 µm aperture, 0.3 N/m stiffness) and later used in an adhesion workflow.
1. Open the blister pack.
2. Place the micropipette on aluminum foil. Make sure that the cantilever is not in contact with the aluminum foil.
3. Fold the aluminum foil, so the probe is closed inside. Make sure it is not too tight, so in case the volume of the pack increases during the sterilization process, it will not break.
4. Seal the sides with autoclave tape.
5. Carefully, place the pack into the autoclave. Start a sterilization program (e.g. for solids, 121°C, 20 min).
6. Once the program is done, let the autoclave and the pack cool down.
7. Transfer the pack to a laminar flow hood. Air dry it ON.
8. Carefully, open the package. The micropipette is ready to use.
About This Community
|Asked: 4/9/19, 8:57 AM|
|Seen: 314 times|
|Last updated: 11/27/19, 3:07 PM|