Welcome to the fluidfm forum
This community is for professionals and enthusiasts of our products and services. Share and discuss the best applications, experiments and new ideas, build your professional profile and become a better researcher together.
Please read the guidelines before participating in this community.
FluidFM probes have contact angles around 9° to 11° depending on the platform.
Bruker BioScope Resolve: 9°
Cytosurge FluidFM BOT: 11°
JPK Nanowizard: 10°
Nanosurf FlexAFM: 10°
For our AFM solutions, we aim to keep the same default angle which the AFM platform uses for all cantilevers.
Keep in mind that this angle will change if the AFM head itself is tilted. Also, there is a small batch-to-batch inherent angle for each probe. Contact us if you need detailed values.
About This Community
|Asked: 7/20/18, 9:23 AM|
|Seen: 469 times|
|Last updated: 11/27/19, 3:07 PM|