- All Products
- FluidFM pneumatic connector
|Dimensions||97 x 6 x 6 mm (LxWxH)|
|Weight||5 g (including tubing)|
|Tubing||Tygon 0.254 / 0.762 mm (inner diameter / outer diameter)|
|O-ring||Viton® DuPont™ 2.5 / 0.65 mm (inner diameter / ring diameter)|
|Glue for PEEK parts||Cyanacrylate with primer|
The FluidFM Pneumatic Connector can be used with:
FluidFM ADD-ON (JPK CellHesion, ForceRobot, NanoWizard and Nanosurf CoreAFM, DriveAFM, FlexAFM)
What is the function of the pneumatic connector?
The FluidFM pneumatic connector guides the pressure from the pressure controller to the probe holder.
How do I clean the pneumatic connector?
To clean your FluidFM pneumatic connector, use only watery solutions or ethanol. Especially avoid strong organic solutions.
How often do I have to exchange the pneumatic connector?
We recommend exchanging the FluidFM pneumatic connector regularly - the latest after 30 experiment days.